Microswitch small – for Polar Bear Water Distillers and Precision Water Distillers
MICROSWITCH, BOILING TANK,PB& PWS,15.1A,(STRAIT LEVER) High Force
Polar Bear Part # 60004
Precision Part # 400007
** Note price is for one switch only. Image shows 2 switches to show details.
This is a universal replacement for Polar Bear residential 26 & 42 model distillers. 26CT, 26D4, 26D8, 42D5, 42D10, 42D25, 100L, 200L
This is a universal replacement for Precision residential model distillers. 8-5, 8-8, 12-12, 12-2, 45-75
This switch may fit other brand of distillers not mentioned above.
Boiler Micro-switch installation
Please read complete instructions before replacing components.
Parts Referred to:
Part# Qty Description
60004 2 Microswitch, Boiling Tank Float
60024 1 Microswitch, Large
Red Robertson Screwdriver
7/16” Deep-Well Socket
Small Adjustable (crescent) Wrench
Ensure that the correct microswitch is included in your replacement kit.
Switches for Polar Bear Water Distillers – 26C6, 26D-8, 26C-8, 42D10, 42D-25 100-L and 200-L should be about 1” long and marked either “V3L-6-D8”, “V-5113D” or AH15379
Switches for Polar Bear Water Distillers – 26-M, 26-E1, and 26-14 should be about 1 3/4” long and labeled “Unimax”
Switches for Precision Water Distillers – 8-5, 8-8, 12-12, 12-20, 45-75 should be about 1” long and marked either “V3L-6-D8”, “V-5113D” or AH15379
Switches for Polar Bear Water Distillers – 8-M should be about 1 3/4” long and labeled “Unimax”
Turn main power switch on distiller off and unplug cord from wall plug.
Drain the boiling tank. CAUTION: DRAIN VALVE MAY BE VERY HOT. Wait at least 1 hour for boiling tank to cool.
Using the Robertson screwdriver, remove the Distiller front cover by removing the 4 screws.
Remove boiling tank lid and lift the float inside the boiler. Note the height of the float when the switch needing replacing is depressed.
Using a 7/16” deep-well socket, remove the nuts attaching the lexan switch plate to the boiling tank.
One wire at a time, disconnect the wires attached to the old micro-switch and connect them to the corresponding post on the new switch. Note the routing of these wires so that they will not interfere with the motion of the float rod or switch arm. Remove the old switch and replace with new switch. Do not fully tighten the kep nuts because the switch position requires adjustment.
Reinstall lexan switch plate on copper studs, again ensuring that wire routing does not interfere with float rod or switch arm motion.
Adjust switch position so that the switch is depressed at the noted float height. Tighten kep nuts and replace front cover. Ensure impurities drain valve is closed.
Leaving the lid off your boiling tank, plug in and turn on your distiller. Check the switch position in the following way: the heater should turn on when the water level is about 1” above the heating element, and the water should shut off when the tank is between 1/3 and 1/2 full. (Precision 8-M and Polar bear 26E1 and 26-M 3/4 full) If your switch is adjusted incorrectly, turn off and unplug your distiller. Remove the front cover and readjust switch position.
Replace front cover and test again.
Replace the boiling tank lid.
The distiller should now operate properly. If it does not, confirm that the wires are connected to the proper location by referring to your electrical schematic.
If they appear correct, we suggest you take your distiller to an authorized service representative, or call 1-780-477-1328 ask for service.
Water Distiller Parts | Health Food Stores Edmonton | Natural Food Stores Edmonton | Supplement Stores Edmonton | Health Supplement Stores Edmonton | Best Supplement Stores Edmonton | Aromatherapy Stores Edmonton | Vitamin Stores Edmonton | Herbal Stores Edmonton | Natural Body Care Edmonton | Cooler Stores Edmonton | Distiller Stores Edmonton | Reverse Osmosis Water Filters Edmonton | Water Pumps Edmonton | Water Pumps for Sale Edmonton